To install click the Add extension button. That's it.

The source code for the WIKI 2 extension is being checked by specialists of the Mozilla Foundation, Google, and Apple. You could also do it yourself at any point in time. 4,5
Kelly Slayton
Congratulations on this excellent venture… what a great idea!
Alexander Grigorievskiy
I use WIKI 2 every day and almost forgot how the original Wikipedia looks like.
Live Statistics
English Articles
Improved in 24 Hours
Languages
Recent
Show all languages
What we do. Every page goes through several hundred of perfecting techniques; in live mode. Quite the same Wikipedia. Just better.
.
Leo
Newton
Brights
Milds Distortionmeter

Distortionmeter (or more precisely distortion factor meter) is an electronic measuring instrument which displays the amount of distortion added to the original signal by an electronic circuit.

Harmonic distortion

Harmonic distortion is equivalent to adding harmonics to a signal. When a purely sinusoidal signal is in this way, a series of harmonics is superimposed on the original signal, and can be detected with suitable equipment.

If the input is

${f_{i}}=a_{1}\cdot sin(\omega t)$ The normalized output is

${f_{o}}=a_{1}\cdot sin(\omega t)+a_{2}\cdot sin(2\omega t)+a_{3}\cdot sin(3\omega t)+..$ The value of Total Harmonics Distortion (THD) is defined as the ratio of the harmonics to the fundamental;  i.e.,

$THD={\frac {\sqrt {{a_{2}}^{2}+{a_{3}}^{2}+..}}{a_{1}}}$ This ratio can be given in dB or in percentage.

The instrument

A distortionmeter is actually a levelmeter with two switchable parallel circuits at the input. The first circuit measures the total signal at the output of a system. (For low distortion levels this will be almost equal to fundamental). That value is adjusted to read 100% or, equivalently, to 0 dB. The second circuit is a high pass filter which removes (as much as practical) the fundamental frequency. This can be a notch filter, one which passes all but the fundamental, with negligible attenuation at other frequencies (including whatever harmonics might be present). Alternatively, if the distortion products are at higher frequencies, a highpass filter can be used if its cutoff rate is sufficiently steep to not affect the expected distortion products. The output of the filter is measured as a percentage of the fundamental, and the reported value will be the distortion value.